||Clicking this icon opens a window that contains specifications and mechanical drawings.
||Clicking this icon allows you to download our standard support documentation.
|Clicking the words "Choose Item" opens a drop-down list containing all of the in-stock lasers around the desired center wavelength. The red icon next to the serial number then allows you to download L-I-V and spectral measurements for that serial-numbered device.
- Single-Wavelength Distributed Feedback Quantum Cascade Lasers (QCLs)
- Typical Output Power from 40 to 120 mW, Depending on Device
- Center Wavelengths from 4.00 µm to 11.00 μm (Wavenumbers Between 2500 cm-1 and 909 cm-1)
- Compact Two-Tab C-Mount Package: 6.4 mm x 4.3 mm x 7.9 mm (L x W x H)
- Electrically Isolated from C-Mount
- Custom Wavelengths and Mounts Also Available (Contact Tech Support for Details)
Thorlabs' Distributed Feedback Quantum Cascade Lasers (QCLs) emit at a well defined center wavelength and provide single spatial mode operation. By tuning the input current and operating temperature, the output frequency can be tuned over a narrow range between 1 cm-1 and 5 cm-1. These lasers are ideal for chemical sensing (see the Spectroscopy tab), optical communications, and other applications. Thorlabs also manufactures Fabry-Perot Quantum Cascade Lasers and Interband Cascade Lasers, which exhibit broadband emission.
Before shipment, the output spectrum, power, and L-I-V curve are measured for each serial-numbered device by an automated test station. These measurements are available below and are also included on a data sheet with the laser. These QCLs are specified for CW output. While pulsed output is possible, this application prohibits current tuning, and performance is not guaranteed. Please note that some optical power is emitted through the rear facet; this output is not usable in applications.
Each laser is mounted on a two-tab C-mount that provides high thermal conductivity and can be secured using a 2-56 or M2 screw with the counterbored Ø2.4 mm (Ø0.09") through hole. As measured from the bottom of the C-mount, the emission height is either 7.15 mm or 7.39 mm depending on the chosen laser. Click on a laser's blue info icon () below and click on the Drawing tab to find the laser's emission height. QCLs are electrically isolated from their C-mounts. Please see the Handling tab for more tips and information for handling these laser packages.
Mounts, Drivers, and Temperature Control
We generally recommend the LDMC20 C-Mount Laser Mount and ITC4002QCL or ITC4005QCL Dual Current / Temperature Controller for these lasers. This device combination includes all the necessary components to mount, drive, and thermally regulate a two-tab C-mount laser. Other compatible current and temperature controllers are listed in the Drivers tab.
If designing your own mounting solution, note that due to these lasers' heat loads, we recommend that they be mounted in a thermally conductive housing to prevent heat buildup. Heat loads for distributed feedback QCLs can be up to 14.2 W (see the Handling tab for additional information).
The typical operating voltage of distributed feedback QCLs is 9 - 14 V. These lasers do not have built-in monitor photodiodes and must be operated in constant current mode.
DFB QCLs at Custom Wavelengths
Thorlabs manufactures custom and OEM quantum cascade lasers in high volumes. We maintain chip inventory from 3 µm to 12 µm at our Jessup, Maryland laser manufacturing facility, and can deliver DFB lasers with custom center wavelengths that are qualified to a user-defined wavelength precision.
More details are available on the Custom & OEM Lasers tab. To inquire about pricing and availability, please contact us. A semiconductor specialist will contact you within 24 hours or the next business day.
|Dual Current / Temperature Controllers
Use the tables below to select a compatible controller for our MIR lasers. The first table lists the controllers with which a particular MIR laser is compatible, and the second table contains selected information on each controller; complete information on each controller is available in its full web presentation. We particularly recommend our ITC4002QCL and ITC4005QCL controllers, which have high compliance voltages of 17 V and 20 V, respectively. Together, these drivers support the current and voltage requirements of our entire line of Mid-IR Lasers. To get L-I-V and spectral measurements of a specific, serial-numbered device, click "Choose Item" next to the part number below, then click on the Docs Icon next to the serial number of the device.
Laser Mount Compatibility
Thorlabs' LDMC20 C-Mount Laser Mount ships with current and TEC cables for the LDC4005, ITC4001, ITC4002QCL, ITC4005, and ITC4005QCL controllers. If designing your own mounting solution, note that due to these lasers' heat loads, we recommend that they be secured in a thermally conductive housing to prevent heat buildup. Heat loads for distributed feedback QCLs can be up to 14.2 W.
Laser and Controller Compatibility
|Laser Item #
||Dual Current / Temperature Controllers
||4.00 - 5.00 µm
(2500 - 2000 cm-1)
||4.54 - 4.62 µm
(2203 - 2165 cm-1)
||ITC4002QCL, ITC4005, ITC4005QCL
||5.00 - 6.00 µm
(2000 - 1667 cm-1)
||5.20 - 5.30 µm
(1923 - 1887 cm-1)
||ITC4002QCL, ITC4005, ITC4005QCL
||6.00 - 7.00 µm
(1667 - 1429 cm-1)
||7.00 - 8.00 µm
(1429 - 1250 cm-1)
||7.90 - 8.00 µm
(1266 - 1250 cm-1)
||ITC4001, ITC4002QCL, ITC4005, ITC4005QCL
||8.00 - 8.10 µm
(1250 - 1235 cm-1)
||ITC4001, ITC4002QCL, ITC4005, ITC4005QCL
||8.00 - 9.00 µm
(1250 - 1111 cm-1)
||8.60 - 8.70 µm
(1163 - 1149 cm-1)
||ITC4001, ITC4002QCL, ITC4005, ITC4005QCL
||9.00 - 10.00 µm
(1111 - 1000 cm-1)
||9.50 - 9.60 µm
(1053 - 1042 cm-1)
||ITC4002QCL, ITC4005, ITC4005QCL
||10.00 - 11.00 µm
(1000 - 909 cm-1)
Controller Selection Guide
|Controller Item #
LDMC20 Laser Mount
(263 x 122 x 307 mm)
|0 to 5 A
||1 mA (Front Panel)
80 µA (Remote Control)
||Included with LDMC20
||Current / Temperature
(263 x 122 x 307 mm)
|0 to 1 A
||100 µA (Front Panel)
16 µA (Remote Control)
||Included with LDMC20
||0 to 2 A
||100 µA (Front Panel)
32 µA (Remote Control)
||Included with LDMC20
||0 to 5 A
||1 mA (Front Panel)
80 µA (Remote Control)
||Included with LDMC20
||Included with LDMC20
- Provide External Temperature Regulation
(e.g., Heat Sinks, Fans, and/or Water Cooling)
- Use a Constant Current Source Specifically Designed for Lasers
- Observe Static Avoidance Practices
- Be Careful When Making Electrical Connections
- Use Thermal Grease
- Expose the Laser to Smoke, Dust, Oils, Adhesive Films, or Flux Fumes
- Blow on the Laser
- Drop the Laser Package
Handling Two-Tab C-Mount Lasers
Proper precautions must be taken when handling and using two-tab C-mount lasers. Otherwise, permanent damage to the device will occur. Members of our Technical Support staff are available to discuss possible operation issues.
Since these lasers are sensitive to electrostatic shock, they should always be handled using standard static avoidance practices.
Avoid Dust and Other Particulates
Unlike TO can and butterfly packages, the laser chip of a two-tab C-mount laser is exposed to air; hence, there is no protection for the delicate laser chip. Contamination of the laser facets must be avoided. Do not blow on the laser or expose it to smoke, dust, oils, or adhesive films. The laser facet is particularly sensitive to dust accumulation. During standard operation, dust can burn onto this facet, which will lead to premature degradation of the laser. If operating a two-tab C-mount laser for long periods of time outside a cleanroom, it should be sealed in a container to prevent dust accumulation.
Use a Current Source Specifically Designed for Lasers
These lasers should always be used with a high-quality constant current driver specifically designed for use with lasers, such as any current controller listed in the Drivers tab. Lab-grade power supplies will not provide the low current noise required for stable operation, nor will they prevent current spikes that result in immediate and permanent damage.
Thermally Regulate the Laser
Temperature regulation is required to operate the laser for any amount of time. The temperature regulation apparatus should be rated to dissipate the maximum heat load that can be drawn by the laser. For our interband cascade lasers, this value is up to 2.5 W, while for our quantum cascade lasers, it can be up to 18 W. The LDMC20 C-Mount Laser Mount, which is compatible with our two-tab C-mount lasers, is rated for >20 W of heat dissipation.
The back face of the C-mount package is machined flat to make proper thermal contact with a heat sink. Ideally, the heat sink will be actively regulated to ensure proper heat conduction. A Thermoelectric Cooler (TEC) is well suited for this task and can easily be incorporated into any standard PID controller.
A fan may serve to move the heat away from the TEC and prevent thermal runaway. However, the fan should not blow air on or at the laser itself. Water cooling methods may also be employed for temperature regulation. Do not use thermal grease with this package, as it can creep, eventually contaminating the laser facet. Pyrolytic graphite is an acceptable alternatives to thermal grease for these packages. Solder can also be used to thermally regulate two-tab C-mount lasers, although controlling the thermal resistance at the interface is important for best results.
Carefully Make Electrical Connections
When making electrical connections, care must be taken. The flux fumes created by soldering can cause laser damage, so care must be taken to avoid this. Solder can be avoided entirely for two-tab C-mount lasers by using the LDMC20 C-Mount Laser Mount. If soldering to the tabs, solder with the C-mount already attached to a heat sink to avoid unnecessary heating of the laser chip.
Minimize Physical Handling
As any interaction with the package carries the risk of contamination and damage, any movement of the laser should be planned in advance and carefully carried out. It is important to avoid mechanical shocks. Dropping the laser package from any height can cause the unit to permanently fail.
Choosing a Collimating Lens
Since the output of our MIR lasers is highly divergent, collimating optics are necessary. Aspheric lenses, which are corrected for spherical aberration, are commonly chosen when the desired beam diameter is between 1 - 5 mm. The simple example below illustrates the key specifications to consider when choosing the correct lens for a given application.
- Center Wavelength: 3.80 µm
- Laser Item #: IF3800CM2
- Desired Collimated Beam Diameter: 4 mm (Major Axis)
The specifications for the IF3800CM2 indicate that the typical parallel and perpendicular FWHM divergences are 40° and 60°, respectively. Therefore, as the light propagates, an elliptical beam will result. To collect as much light as possible during the collimation process, consider the larger of these two divergence angles in your calculations (in this case, 60°).
θ = Divergence Angle
Ø = Beam Diameter
Using the information above, the focal length needed to obtain the desired beam diameter can be calculated:
This information allows the appropriate collimating lens to be selected. Thorlabs offers a large selection of black diamond aspheric lenses for the mid-IR spectral range. Since this laser emits at 3.80 µm, the best AR coating is our -E coating, which provides Ravg < 0.6% per surface from 3 to 5 µm. The lenses with focal lengths closest to the calculated value of 3.46 mm are our 390036-E (unmounted) or C036TME-E (mounted) Molded Aspheric Lenses, which have f = 4.00 mm. Plugging this focal length back into the equation shown above gives a final beam diameter of 4.62 mm along the major axis.
Next, we verify that the numerical aperture (NA) of the lens is larger than the NA of the laser:
NALens = 0.56
NALaser ~ sin (30°) = 0.5
NALens > NALaser
Since NALens > NALaser, the 390036-E or C036TME-E lenses will give acceptable beam quality. However, by using the FWHM beam diameter, we have not accounted for a significant fraction of the beam power. A better practice is to use the 1/e2 beam diameter. For a Gaussian beam profile, the 1/e2 beam diameter is approximately equal to 1.7X the FWHM diameter. The 1/e2 beam diameter is therefore a more conservative estimate of the beam size, containing more of the laser's intensity. Using this value significantly reduces far-field diffraction (since less of the incident light is clipped) and increases the power delivered after the lens.
A good rule of thumb is to pick a lens with an NA of twice the NA of the laser diode. For example, either the 390037-E or the C037TME-E could be used as these lenses each have an NA of 0.85, which a little less than twice that of our IF3800CM2 laser (NA 0.5). Compared to the first set of lenses we identified, these have a shorter focal length of 1.873 mm, resulting in a smaller final beam diameter of 2.16 mm.
Beam Profile Characterization of a Mid-IR Laser
Because quantum cascade lasers (QCLs) and interband cascade lasers (ICLs) have intrinsically large divergence angles, it is necessary to install collimating optics in front of the laser face, as shown in the Collimation tab. We are frequently asked what beam quality can be reasonably expected once the beam has been collimated. This tab presents an M2 measurement we performed using our 3.80 µm Interband Cascade Laser (Item # IF3800CM2).
For this system, we obtained M2 = 1.2 ± 0.08 in the parallel direction and M2 = 1.3 ± 0.2 in the perpendicular direction. While this is just one example, we believe these results to be representative of well-collimated mid-IR lasers manufactured by Thorlabs, as corroborated by supplementary measurements we have performed in-house.
The apparatus we used to determine M2 is shown schematically in the figure above. In order to ensure that our results were rigorous, all data acquisition and analysis were consistent with the ISO11146 standard.
The IF3800CM2 Interband Cascade Laser used for this measurement emitted CW laser light with a center wavelength of 3.781 µm. Our LDMC20 temperature-stabilized mount held the laser's temperature at 25 °C. The output beam was collimated by a C037TME-E lens located immediately downstream of the laser face. This lens was selected because of its large NA of 0.85 (which helped maximize collection of the emitted light) and because of its AR coating (Ravg < 0.6% per surface from 3 µm to 5 µm). We measured 10 mW of output power after the lens.
A pyroelectric camera (Spiricon Pyrocam IV) with 80 µm square pixels was scanned along the beam propagation direction, and the beam width was measured along the parallel and perpendicular directions using the second-order moment (D4σ) definition. Hyperbolas were fit to the beam width to extract M2 for each direction. The camera's internal chopper was triggered at 50 Hz since the pyroelectric effect is sensitive to changes in temperature rather than absolute temperature differences. A ZnSe window was present in front of the detector array to help minimize visible light contributions to the signal.
Presented to the right are the second-order moment (D4σ) beam widths for the parallel and perpendicular directions as a function of distance from the laser face (denoted as z). Along the parallel direction, we obtained a minimum beam width of 1.5 mm, while along the perpendicular direction, we obtained a minimum beam width of 1.3 mm. The spatial profiles we observed at the two minimum beam width positions, as obtained by the pyroelectric camera, are shown below.
The divergence of the beam can be described by a hyperbola, as written in Equation 1:
In order to obtain the hyperbola coefficients a, b, and c for the parallel and perpendicular directions, we fit the discrete beam width measurements along each direction to hyperbolas, as shown in the graph to the right. These coefficients were substituted into Equation 2 (taking λ = 3.781 µm) to yield M2.
The hyperbola coefficients and M2 values derived by this method are listed in the table below.
||3.6 ± 0.1 mm2
||9.7 ± 0.2 mm2
||-0.0096 ± 0.0007 mm
||-0.0268 ± 0.0008 mm
||(1.61 ± 0.08) × 10-5
||(2.27 ± 0.08) × 10-5
||1.2 ± 0.08
||1.3 ± 0.2
The 0.85 NA of the collimating lens we used is the largest NA of any lens for this wavelength range that is offered in our catalog. Despite this large NA, we observed lobes in the far field (shown by the figure below) that are consistent with clipping of the laser-emitted light. An ideal measurement would not contain these artifacts.
As shown by the graph above and to the right, we observed significant astigmatism in the collimated beam: the beam waist of the parallel direction occurred around z = 300 mm, while the beam waist of the perpendicular direction occurred around z = 600 mm. This astigmatism corresponds closely to what is expected for this laser, given that the IF3800CM2 laser is specified with a parallel FWHM beam divergence of 40° and a perpendicular FWHM beam divergence of 60°.
Beam Profile at Beam Waist of Parallel Direction
(Each Pixel is 80 µm Square)
Beam Profile at Beam Waist of Perpendicular Direction
(Each Pixel is 80 µm Square)
|Selected Distributed Feedback QCLsa|
||Nominal Center Frequency
||1047 cm-1 (9.55 µm)
||1156 cm-1 (8.65 µm)
SO2 (Sulfur Dioxide)
||1242 cm-1 (8.05 µm)
HONO (Nitrous Acid)
||1258 cm-1 (7.95 µm)
HONO (Nitrous Acid)
||1905 cm-1 (5.25 µm)
||NO (Nitric Oxide)
||2183 cm-1 (4.58 µm)
||CO (Carbon Monoxide)
N2O (Nitrous Oxide)
Gas-Phase Spectroscopy Using Distributed Feedback Lasers
Distributed Feedback Quantum Cascade Lasers (DFB QCLs) offer many attractive features for spectroscopy. They emit at a single wavelength within the mid-IR, where many gaseous species characteristically absorb. Moreover, their emission wavelength is easily tuned (typical tuning range: 1 - 5 cm-1) by changing the drive current and operating temperature of the laser, making them ideal for isolating narrow gas absorption lines. Finally, they offer relatively high output power (typically 40 - 120 mW at rollover current), helping improve measurement sensitivity.
Thorlabs' DFB QCLs emit at wavelengths that range from 4.00 to 11.00 µm (2500 cm-1 to 909 cm-1). If we do not stock the wavelength required for your application, custom wavelengths are available by contacting Tech Support.
The tuning range of individual DFB QCLs depends greatly on the actual laser device. Each DFB QCL is a unique device with its own threshold current, rollover current, and spectrum. With typical lasers, it is usually preferable to operate the laser at or near the rollover current, since the output power is lowest at threshold and highest at rollover. On the other hand, the wavelength of DFB QCLs changes as a function of the current, so operating at the rollover current is not always possible in spectroscopy measurements, which require specific wavelengths. (It is important to note that the output power is not constant over the entire tuning range.)
To demonstrate DFB QCLs' tunability, we measured the center wavelength of a QD4580CM1 laser as a function of drive current, from threshold to near rollover, at 15 °C and 25 °C. Over the entire temperature and drive current range, we obtained center wavelengths from 4.568 µm to 4.586 µm (2189.14 cm-1 to 2180.77 cm-1), spanning a range of 18 nm (8.37 cm-1), with output power ranging from 3.2 mW (at threshold current) to 39.1 mW (at near-rollover current). Since the laser is capable of operating at 35 °C, even broader wavelength tuning is also achievable.
Click to Enlarge
DFB QCL Center Frequency as Function of Temperature and Drive Current
|Sample QD4580CM1 Spectrum and Output Power|
||2189.14 cm-1 (4.568 µm)
||2187.34 cm-1 (4.572 µm)
||2188.12 cm-1 (4.570 µm)
||2186.26 cm-1 (4.574 µm)
||2186.92 cm-1 (4.573 µm)
||2185.05 cm-1 (4.577 µm)
||2185.71 cm-1 (4.575 µm)
||2183.78 cm-1 (4.579 µm)
||2184.33 cm-1 (4.578 µm)
||2182.34 cm-1 (4.582 µm)
||2182.76 cm-1 (4.581 µm)
||2180.77 cm-1 (4.586 µm)
ECL, DFB, VHG-Stabilized, and DBR Single-Frequency Lasers
A wide variety of applications require tunable single-frequency operation of a laser system. In the world of diode lasers, there are currently four main configurations to obtain a single-frequency output: external cavity laser (ECL), distributed feedback (DFB), volume holographic grating (VHG), and distributed Bragg reflector (DBR). All four are capable of single-frequency output through the utilization of grating feedback. However, each type of laser uses a different grating feedback configuration, which influences performance characteristics such as output power, tuning range, and side mode suppression ratio (SMSR). We discuss below some of the main differences between these four types of single-frequency diode lasers.
External Cavity Laser
The External Cavity Laser (ECL) is a versatile configuration that is compatible with most standard free space diode lasers. This means that the ECL can be used at a variety of wavelengths, dependent upon the internal laser diode gain element. A lens collimates the output of the diode, which is then incident upon a grating (see Figure 1). The grating provides optical feedback and is used to select the stabilized output wavelength. With proper optical design, the external cavity allows only a single longitudinal mode to lase, providing single-frequency laser output with high side mode suppression ratio (SMSR > 45 dB).
One of the main advantages of the ECL is that the relatively long cavity provides extremely narrow linewidths (<1 MHz). Additionally, since it can incorporate a variety of laser diodes, it remains one of the few configurations that can provide narrow linewidth emission at blue or red wavelengths. The ECL can have a large tuning range (>100 nm) but is often prone to mode hops, which are very dependent on the ECL's mechanical design as well as the quality of the antireflection (AR) coating on the laser diode.
Click to EnlargeFigure 2:
DFB Lasers Have a Bragg Reflector Along the Length of the Active Gain Medium
Distributed Feedback Laser
The Distributed Feedback (DFB) Laser (available in NIR and MIR) incorporates the grating within the laser diode structure itself (see Figure 2). This corrugated periodic structure coupled closely to the active region acts as a Bragg reflector, selecting a single longitudinal mode as the lasing mode. If the active region has enough gain at frequencies near the Bragg frequency, an end reflector is unnecessary, relying instead upon the Bragg reflector for all optical feedback and mode selection. Due to this “built-in” selection, a DFB can achieve single-frequency operation over broad temperature and current ranges. To aid in mode selection and improve manufacturing yield, DFB lasers often utilize a phase shift section within the diode structure as well.
The lasing wavelength for a DFB is approximately equal to the Bragg wavelength:
where λ is the wavelength, neff is the effective refractive index, and Λ is the grating period. By changing the effective index, the lasing wavelength can be tuned. This is accomplished through temperature and current tuning of the DFB.
The DFB has a relatively narrow tuning range: about 2 nm at 850 nm, about 4 nm at 1550 nm, or at least 1 cm-1 in the mid-IR (4.00 - 11.00 µm). However, over this tuning range, the DFB can achieve single-frequency operation, which means that this is a continuous tuning range without mode hops. Because of this feature, DFBs have become a popular and majority choice for real-world applications such as telecom and sensors. Since the cavity length of a DFB is rather short, the linewidths are typically in the 1 MHz to 10 MHz range. Additionally, the close coupling between the grating structure and the active region results in lower maximum output power compared to ECL and DBR lasers.
Click to EnlargeFigure 3:
VHG Lasers have a Volume Holographic Grating Outside of the Active Gain Medium
A Volume-Holographic-Grating-(VHG)-Stabilized Laser also uses a Bragg reflector, but in this case a transmission grating is placed in front of the laser diode output (see Figure 3). Since the grating is not part of the laser diode structure, it can be thermally decoupled from the laser diode, improving the wavelength stability of the device. The grating typically consists of a piece of photorefractive material (typically glass) which has a periodic variation in the index of refraction. Only the wavelength of light that satisfies the Bragg condition for the grating is reflected back into the laser cavity, which results in a laser with extremely wavelength-stable emission. A VHG-Stabilized laser can produce output with a similar linewidth to a DFB laser at higher powers that is wavelength-locked over a wide range of currents and temperatures.
Click to EnlargeFigure 4:
DBR Lasers have a Bragg Reflector Outside of the Active Gain Medium
Distributed Bragg Reflector Laser
Similar to DFBs, Distributed Bragg Reflector (DBR) lasers incorporate an internal grating structure. However, whereas DFB lasers incorporate the grating structure continuously along the active region (gain region), DBR lasers place the grating structure(s) outside this region (see Figure 4). In general a DBR can incorporate various regions not typically found in a DFB that yield greater control and tuning range. For instance, a multiple-electrode DBR laser can include a phase-controlled region that allows the user to independently tune the phase apart from the grating period and laser diode current. When utilized together, the DBR can provide single-frequency operation over a broad tuning range. For example, high end sample-grating DBR lasers can have a tuning range as large as 30 - 40 nm. Unlike the DFB, the output is not mode hop free; hence, careful control of all inputs and temperature must be maintained.
In contrast to the complicated control structure for the multiple-electrode DBR, a simplified version of the DBR is engineered with just one electrode. This single-electrode DBR eliminates the complications of grating and phase control at the cost of tuning range. For this architecture type, the tuning range is similar to a DFB laser but will mode hop as a function of the applied current and temperature. Despite the disadvantage of mode hops, the single-electrode DBR does provide some advantages over its DFB cousin, namely higher output power because the grating is not continuous along the length of the device. Both DBR and DFB lasers have similar laser linewidths. Currently, Thorlabs offers only single-electrode DBR lasers.
ECL, DFB, VHG, and DBR laser diodes provide single-frequency operation over their designed tuning range. The ECL can be designed for a larger selection of wavelengths than either the DFB or DBR. While prone to mode hops, it also provides the narrowest linewidth (<1 MHz) of the three choices. In appropriately designed instruments, ECLs can also provide extremely broad tuning ranges (>100 nm).
The DFB laser is the most stable single-frequency, tunable laser of the four. It can provide mode-hop-free performance over its entire tuning range, making it one of the most popular forms of single-frequency laser for much of industry. It has the lowest output power due to inherent properties of the continuous grating feedback structure.
The VHG laser provides the most stable wavelength performance over a range of temperatures and currents and can provide higher powers than are typical in DFB lasers. This stability makes it excellent for use in OEM applications.
The single-electrode DBR laser provides similar linewidth and tuning range as the DFB (<5 nm). However, the single-electrode DBR will have periodic mode hops in its tuning curve.
Custom & OEM Quantum Cascade and Interband Cascade Lasers
At our semiconductor manufacturing facility in Jessup, Maryland, we build a wide range of mid-IR lasers and gain chips. Our engineering team performs in-house epitaxial growth, wafer fabrication, and laser packaging. We maintain chip inventory from 3 µm to 12 µm, and our vertically integrated facilities are well equipped to fulfill unique requests.
High-Power Fabry-Perot QCLs
For Fabry-Perot lasers, we can reach multi-watt output power on certain custom orders. The available power depends upon several factors, including the wavelength and the desired package.
DFB QCLs at Custom Wavelengths
For distributed feedback (DFB) lasers, we can deliver a wide range of center wavelengths with user-defined wavelength precision. Our semiconductor specialists will take your application requirements into account when discussing the options with you.
The graphs below and photos to the right illustrate some of our custom capabilities. Please visit our semiconductor manufacturing capabilities presentation to learn more.
|The rows shaded green below denote single-frequency lasers.|
|L375P70MLD||375 nm||70 mW||110 mA||5.4 V||9°||22.5°||Single Mode||Ø5.6 mm|
|L404P400M||404 nm||400 mW||370 mA||4.9 V||13° (1/e2)||42° (1/e2)||Multimode||Ø5.6 mm|
|LP405-SF10||405 nm||10 mW||50 mA||5.0 V||-||-||Single Mode||Ø5.6 mm, SM Pigtail|
|L405P20||405 nm||20 mW||38 mA||4.8 V||8.5°||19°||Single Mode||Ø5.6 mm|
|LP405-SF30||405 nm||30 mW||100 mA||4.8 V||-||-||Single Mode||Ø5.6 mm, SM Pigtail|
|DL5146-101S||405 nm||40 mW||70 mA||5.2 V||8°||19°||Single Mode||Ø5.6 mm|
|LP405-MF80||405 nm||80 mW||85 mA||5.0 V||-||-||Multimode||Ø5.6 mm, MM Pigtail|
|LP405-MF300||405 nm||300 mW||350 mA||4.5 V||-||-||Multimode||Ø5.6 mm, MM Pigtail|
|LP406-SF20||406 nm||20 mW||75 mA||4.8 V||-||-||Single Mode||Ø5.6 mm, SM Pigtail|
|LP450-SF15||450 nm||15 mW||85 mA||5.5 V||-||-||Single Mode||Ø9 mm, SM Pigtail|
|PL450B||450 nm||80 mW||100 mA||5.8 V||4 - 11°||18 - 25°||Single Mode||Ø3.8 mm|
|L462P1400MM||462 nm||1400 mW||1350 mA||5 V||5 - 25°||35 - 50°||Multimode||Ø9 mm|
|LP462-MF1W||462 nm||1000 mW||1100 mA||5.0 V||-||-||Multimode||Ø9 mm, MM Pigtail|
|LP473-SF6||473 nm||6 mW||80 mA||5.0 V||-||-||Single Mode||Ø5.6 mm, SM Pigtail|
|LP488-SF20||488 nm||20 mW||70 mA||6.0 V||-||-||Single Mode||Ø5.6 mm, SM Pigtail|
|L488P60||488 nm||60 mW||75 mA||6.8 V||7||23||Single Mode||Ø5.6 mm|
|LP520-SF15||520 nm||15 mW||140 mA||6.5 V||-||-||Single Mode||Ø9 mm, SM Pigtail|
|L520P50||520 nm||45 mW||150 mA||7.0 V||7°||22°||Single Mode||Ø5.6 mm|
|PL520||520 nm||50 mW||150 mA||7.0 V||7°||22°||Single Mode||Ø3.8 mm|
|LP520-MF100||520 nm||100 mW||320 mA||6.0 V||-||-||Multimode||Ø5.6 mm, MM Pigtail|
|DJ532-10||532 nm||10 mW||220 mA||1.9 V||0.69°||0.69°||Single Mode||Ø9.5 mm (non-standard)|
|DJ532-40||532 nm||40 mW||330 mA||1.9 V||0.69°||0.69°||Single Mode||Ø9.5 mm (non-standard)|
|HL63163DG||633 nm||100 mW||170 mA||2.6 V||8.5°||18°||Single Mode||Ø5.6 mm|
|LPS-635-FC||635 nm||2.5 mW||70 mA||2.2 V||-||-||Single Mode||Ø9.5 mm, SM Pigtail|
|LPS-PM635-FC||635 nm||2.5 mW||70 mA||2.2 V||-||-||Single Mode||Ø9.5 mm, PM Pigtail|
|L635P5||635 nm||5 mW||30 mA||<2.7 V||8°||32°||Single Mode||Ø5.6 mm|
|HL6312G||635 nm||5 mW||55 mA||<2.7 V||8°||31°||Single Mode||Ø9 mm|
|LPM-635-SMA||635 nm||8 mW||50 mA||2.2 V||-||-||Multimode||Ø9 mm, MM Pigtail|
|LP635-SF8||635 nm||8 mW||60 mA||2.3 V||-||-||Single Mode||Ø5.6 mm, SM Pigtail|
|HL6320G||635 nm||10 mW||70 mA||<2.7 V||8°||31°||Single Mode||Ø9 mm|
|HL6322G||635 nm||15 mW||85 mA||<2.7 V||8°||30°||Single Mode||Ø9 mm|
|L637P5||637 nm||5 mW||20 mA||<2.4 V||8°||34°||Single Mode||Ø5.6 mm|
|LP637-SF70||637 nm||70 mW||220 mA||2.7 V||-||-||Single Mode||Ø5.6 mm, SM Pigtail|
|HL63142DG||637 nm||100 mW||140 mA||2.7 V||8°||18°||Single Mode||Ø5.6 mm|
|HL63133DG||637 nm||170 mW||250 mA||2.8 V||9°||17°||Single Mode||Ø5.6 mm|
|HL6388MG||637 nm||250 mW||340 mA||2.3 V||10°||40°||Multimode||Ø5.6 mm|
|L638P040||638 nm||40 mW||92 mA||2.4 V||10°||21°||Single Mode||Ø5.6 mm|
|L638P700M||638 nm||700 mW||820 mA||2.2 V||9°||35°||Multimode||Ø5.6 mm|
|HL6358MG||639 nm||10 mW||40 mA||2.3 V||8°||21°||Single Mode||Ø5.6 mm|
|HL6323MG||639 nm||30 mW||95 mA||2.3 V||8.5°||30°||Single Mode||Ø5.6 mm|
|HL6362MG||640 nm||40 mW||90 mA||2.4 V||10°||21°||Single Mode||Ø5.6 mm|
|LP642-SF20||642 nm||20 mW||90 mA||2.5 V||-||-||Single Mode||Ø5.6 mm, SM Pigtail|
|LP642-PF20||642 nm||20 mW||90 mA||2.5 V||-||-||Single Mode||Ø5.6 mm, PM Pigtail|
|HL6364DG||642 nm||60 mW||125 mA||2.5 V||10°||21°||Single Mode||Ø5.6 mm|
|HL6366DG||642 nm||80 mW||155 mA||2.5 V||10°||21°||Single Mode||Ø5.6 mm|
|HL6385DG||642 nm||150 mW||280 mA||2.6 V||9°||17°||Single Mode||Ø5.6 mm|
|L650P007||650 nm||7 mW||28 mA||2.2 V||9°||28°||Single Mode||Ø5.6 mm|
|LPS-660-FC||658 nm||7.5 mW||65 mA||2.6 V||-||-||Single Mode||Ø5.6 mm, SM Pigtail|
|LP660-SF20||658 nm||20 mW||80 mA||2.6 V||-||-||Single Mode||Ø5.6 mm, SM Pigtail|
|LPM-660-SMA||658 nm||22.5 mW||65 mA||2.6 V||-||-||Multimode||Ø5.6 mm, MM Pigtail|
|HL6501MG||658 nm||30 mW||65 mA||2.6 V||8.5°||22°||Single Mode||Ø5.6 mm|
|L658P040||658 nm||40 mW||75 mA||2.2 V||10°||20°||Single Mode||Ø5.6 mm|
|LP660-SF40||658 nm||40 mW||135 mA||2.5 V||-||-||Single Mode||Ø5.6 mm, SM Pigtail|
|LP660-SF60||658 nm||60 mW||210 mA||2.4 V||-||-||Single Mode||Ø5.6 mm, SM Pigtail|
|HL6544FM||660 nm||50 mW||115 mA||2.3 V||10°||17°||Single Mode||Ø5.6 mm|
|HL6545MG||660 nm||120 mW||170 mA||2.45 V||10°||17°||Single Mode||Ø5.6 mm|
|L660P120||660 nm||120 mW||175 mA||2.5 V||10°||17°||Single Mode||Ø5.6 mm|
|LPS-675-FC||670 nm||2.5 mW||55 mA||2.2 V||-||-||Single Mode||Ø9 mm, SM Pigtail|
|HL6748MG||670 nm||10 mW||30 mA||2.2 V||8°||25°||Single Mode||Ø5.6 mm|
|HL6714G||670 nm||10 mW||55 mA||<2.7 V||8°||22°||Single Mode||Ø9 mm|
|HL6756MG||670 nm||15 mW||35 mA||2.3 V||8°||24°||Single Mode||Ø5.6 mm|
|SLD1332V||670 nm||500 mW||800 mA||2.4 V||8°||23°||Multimode||Ø9 mm|
|LP685-SF15||685 nm||15 mW||55 mA||2.1 V||-||-||Single Mode||Ø5.6 mm, SM Pigtail|
|HL6750MG||685 nm||50 mW||75 mA||2.3 V||9°||21°||Single Mode||Ø5.6 mm|
|HL6738MG||690 nm||30 mW||90 mA||2.5 V||8.5°||19°||Single Mode||Ø5.6 mm|
|LP705-SF15||705 nm||15 mW||55 mA||2.3 V||-||-||Single Mode||Ø5.6 mm, SM Pigtail|
|HL7001MG||705 nm||40 mW||75 mA||2.5 V||9°||18°||Single Mode||Ø5.6 mm|
|HL7302MG||730 nm||40 mW||75 mA||2.5 V||9°||18°||Single Mode||Ø5.6 mm|
|L780P010||780 nm||10 mW||24 mA||1.8 V||8°||30°||Single Mode||Ø5.6 mm|
|LP780-SAD15||780 nm||15 mW||180 mA||2.2 V||-||-||Single Frequency||Ø9 mm, SM Pigtail|
|L785P5||785 nm||5 mW||28 mA||1.9 V||10°||29°||Single Mode||Ø5.6 mm|
|LPS-PM785-FC||785 nm||6.25 mW||65 mA||-||-||-||Single Mode||Ø5.6 mm, PM Pigtail|
|LPS-785-FC||785 nm||10 mW||65 mA||1.85 V||-||-||Single Mode||Ø5.6 mm, SM Pigtail|
|LP785-SF20||785 nm||20 mW||85 mA||1.9 V||-||-||Single Mode||Ø5.6 mm, SM Pigtail|
|DBR785S||785 nm||22 mW||230 mA||2.0 V||-||-||Single Frequency||SM, Butterfly|
|DBR785P||785 nm||22 mW||230 mA||2.0 V||-||-||Single Frequency||PM, Butterfly|
|L785P25||785 nm||25 mW||45 mA||1.9 V||8°||30°||Single Mode||Ø5.6 mm|
|LP785-SAV50||785 nm||50 mW||500 mA||2.2 V||-||-||Single Frequency||Ø9 mm, SM Pigtail|
|L785P090||785 nm||90 mW||120 mA||2.0 V||9°||16°||Single Mode||Ø5.6 mm|
|LP785-SF100||785 nm||100 mW||300 mA||2.0 V||-||-||Single Mode||Ø9 mm, SM Pigtail|
|FPL785S-250||785 nm||250 mW (Min)||500 mA||2.0 V||-||-||Single Mode||SM Butterfly|
|FPL785P-200||785 nm||200 mW (Min)||450 mA||2.0 V||-||-||Single Mode||PM Butterfly|
|LD785-SEV300||785 nm||300 mW||500 mA (Max)||2.0 V||8°||16°||Single Frequency||Ø9 mm|
|LD785-SH300||785 nm||300 mW||400 mA||2.0 V||7°||18°||Single Mode||Ø9 mm|
|FPL785C||785 nm||300 mW||400 mA||2.0 V||7°||18°||Single Mode||3 mm x 5 mm Submount|
|FPL785CM||785 nm||300 mW||400 mA||2.0 V||7°||18°||Single Mode||C-Mount|
|LD785-SE400||785 nm||400 mW||550 mA||2.0 V||7°||16°||Single Mode||Ø9 mm|
|ML620G40||805 nm||500 mW||650 mA||1.9 V||3°||34°||Multimode||Ø5.6 mm|
|L808P010||808 nm||10 mW||50 mA||2 V||10°||30°||Single Mode||Ø5.6 mm|
|LP808-SF30||808 nm||30 mW||110 mA||1.8 V||-||-||Single Mode||Ø5.6 mm, SM Pigtail|
|L808P030||808 nm||30 mW||65 mA||2 V||10°||30°||Single Mode||Ø5.6 mm|
|LP808-SA40||808 nm||40 mW||140 mA||2.0 V||-||-||Single Mode||Ø9 mm, SM Pigtail|
|LD808-SA60||808 nm||60 mW||100 mA||2.0 V||8°||24°||Single Mode||Ø5.6 mm|
|LD808-SA100||808 nm||100 mW||145 mA||2.0 V||8°||24°||Single Mode||Ø9 mm|
|M9-808-0150||808 nm||150 mW||180 mA||1.9 V||8°||17°||Single Mode||Ø9 mm|
|L808P200||808 nm||200 mW||260 mA||2 V||10°||30°||Multimode||Ø5.6 mm|
|LD808-SEV500||808 nm||500 mW||800 mA (Max)||2.2 V||8°||14°||Single Frequency||Ø9 mm|
|LD808-SE500||808 nm||500 mW||750 mA||2.2 V||7°||14°||Single Mode||Ø9 mm|
|L808P500MM||808 nm||500 mW||650 mA||1.8 V||12°||30°||Multimode||Ø5.6 mm|
|L808P1000MM||808 nm||1000 mW||1100 mA||2 V||9°||30°||Multimode||Ø9 mm|
|L820P100||820 nm||100 mW||145 mA||2.1 V||9°||17°||Single Mode||Ø5.6 mm|
|L820P200||820 nm||200 mW||250 mA||2.4 V||9°||17°||Single Mode||Ø5.6 mm|
|LPS-830-FC||830 nm||10 mW||120 mA||-||-||-||Single Mode||Ø5.6 mm, SM Pigtail|
|LPS-PM830-FC||830 nm||10 mW||120 mA||-||-||-||Single Mode||Ø5.6 mm, PM Pigtail|
|LP830-SF30||830 nm||30 mW||115 mA||1.9 V||-||-||Single Mode||Ø9 mm, SM Pigtail|
|HL8338MG||830 nm||50 mW||75 mA||1.9 V||9°||22°||Single Mode||Ø5.6 mm|
|LD830-SE650||830 nm||650 mW||900 mA||2.3 V||7°||13°||Single Mode||Ø9 mm|
|LD830-MA1W||830 nm||1 W||1.330 A||2.1 V||7°||24°||Multimode||Ø9 mm|
|LD830-ME2W||830 nm||2 W||3 A (Max)||2.0 V||8°||21°||Multimode||Ø9 mm|
|L850P010||850 nm||10 mW||50 mA||2 V||10°||30°||Single Mode||Ø5.6 mm|
|L850P030||850 nm||30 mW||65 mA||2 V||8.5°||30°||Single Mode||Ø5.6 mm|
|LP852-SF30||852 nm||30 mW||115 mA||1.9 V||-||-||Single Mode||Ø9 mm, SM Pigtail|
|DBR852S||852 nm||40 mW||140 mA||2.0 V||-||-||Single Frequency||SM, Butterfly|
|DBR852P||852 nm||40 mW||140 mA||2.0 V||-||-||Single Frequency||PM, Butterfly|
|L852P50||852 nm||50 mW||75 mA||1.9 V||9°||22°||Single Mode||Ø5.6 mm|
|L852P100||852 nm||100 mW||120 mA||1.9 V||8°||18°||Single Mode||Ø9 mm|
|L852P150||852 nm||150 mW||170 mA||1.9 V||8°||18°||Single Mode||Ø9 mm|
|LD852-SE600||852 nm||600 mW||950 mA||2.3 V||7° (1/e2)||13° (1/e2)||Single Mode||Ø9 mm|
|LD852-SEV600||852 nm||600 mW||1050 mA (Max)||2.2 V||8°||13° (1/e2)||Single Frequency||Ø9 mm|
|LP880-SF3||880 nm||3 mW||25 mA||2.2 V||-||-||Single Mode||Ø5.6 mm, SM Pigtail|
|L880P010||880 nm||10 mW||30 mA||2.0 V||12°||37°||Single Mode||Ø5.6 mm|
|L904P010||904 nm||10 mW||50 mA||2 V||10°||30°||Single Mode||Ø5.6 mm|
|LP915-SF40||915 nm||40 mW||130 mA||1.5 V||-||-||Single Mode||Ø9 mm, SM Pigtail|
|M9-915-0200||915 nm||200 mW||260 mA||1.9 V||8°||28°||Single Mode||Ø9 mm|
|M9-915-0300||915 nm||300 mW||370 mA||1.9 V||8°||28°||Single Mode||Ø9 mm|
|LP940-SF30||940 nm||30 mW||90 mA||1.5 V||-||-||Single Mode||Ø9 mm, SM Pigtail|
|M9-940-0100||940 nm||100 mW||140 mA||1.9 V||8°||28°||Single Mode||Ø9 mm|
|M9-940-0200||940 nm||200 mW||270 mA||1.9 V||8°||28°||Single Mode||Ø9 mm|
|DBR976S||976 nm||45 mW||150 mA||2.0 V||-||-||Single Frequency||SM, Butterfly|
|DBR976P||976 nm||45 mW||150 mA||2.0 V||-||-||Single Frequency||PM, Butterfly|
|BL976-SAG300||976 nm||300 mW||470 mA||2.0 V||-||-||Single Mode||SM, Butterfly|
|BL976-PAG500||976 nm||500 mW||830 mA||2.0 V||-||-||Single Mode||PM, Butterfly|
|BL976-PAG700||976 nm||700 mW||1090 mA||2.0 V||-||-||Single Mode||PM, Butterfly|
|BL976-PAG900||976 nm||900 mW||1480 mA||2.5 V||-||-||Single Mode||PM, Butterfly|
|L980P010||980 nm||10 mW||25 mA||2 V||10°||30°||Single Mode||Ø5.6 mm|
|L980P030||980 nm||30 mW||100 mA||1.5 V||10°||30°||Single Mode||Ø5.6 mm|
|L9805E2P5||980 nm||50 mW||95 mA||1.5 V||8°||33°||Single Mode||Ø5.6 mm|
|L980P100A||980 nm||100 mW||150 mA||1.6 V||6°||32°||Multimode||Ø5.6 mm|
|L980P200||980 nm||200 mW||300 mA||1.5 V||6°||30°||Multimode||Ø5.6 mm|
|L1060P200J||1060 nm||200 mW||280 mA||1.3 V||6°||32°||Single Mode||Ø9 mm|
|DBR1064S||1064 nm||20 mW||150 mA||2.0 V||-||-||Single Frequency||SM, Butterfly|
|DBR1064P||1064 nm||20 mW||150 mA||2.0 V||-||-||Single Frequency||PM, Butterfly|
|LPS-1060-FC||1064 nm||50 mW||220 mA||1.4 V||-||-||Single Mode||Ø9 mm, SM Pigtail|
|M9-A64-0200||1064 nm||200 mW||280 mA||1.7 V||8°||28°||Single Mode||Ø9 mm|
|M9-A64-0300||1064 nm||300 mW||390 mA||1.7 V||8°||28°||Single Mode||Ø9 mm|
|BAL1112CM||1208 nm||3000 mW||5000 mA||1.33 V||20°||26°||Multimode||C-Mount|
|LP1310-SAD2||1310 nm||2.0 mW||40 mA||1.1 V||-||-||Single Frequency||Ø5.6 mm, SM Pigtail|
|LPS-1310-FC||1310 nm||2.5 mW||20 mA||1.1 V||-||-||Single Mode||Ø5.6 mm, SM Pigtail|
|LPS-PM1310-FC||1310 nm||2.5 mW||20 mA||1.1 V||-||-||Single Mode||Ø5.6 mm, PM Pigtail|
|L1310P5DFB||1310 nm||5 mW||20 mA||1.1 V||7°||9°||Single Frequency||Ø5.6 mm|
|ML725B8F||1310 nm||5 mW||20 mA||1.1 V||25°||30°||Single Mode||Ø5.6 mm|
|LPSC-1310-FC||1310 nm||50 mW||350 mA||2 V||-||-||Single Mode||Ø5.6 mm, SM Pigtail|
|FPL1053S||1310 nm||130 mW||400 mA||1.7 V||-||-||Single Mode||SM Butterfly|
|FPL1053P||1310 nm||130 mW||400 mA||1.7 V||-||-||Single Mode||PM Butterfly|
|FPL1053T||1310 nm||300 mW (Pulsed)||750 mA||2 V||15°||28°||Single Mode||Ø5.6 mm|
|FPL1053C||1310 nm||300 mW (Pulsed)||750 mA||2 V||15°||27°||Single Mode||Chip on Submount|
|LPS-1550-FC||1550 nm||1.5 mW||30 mA||1.0 V||-||-||Single Mode||Ø5.6 mm, SM Pigtail|
|LPS-PM1550-FC||1550 nm||1.5 mW||30 mA||1.1 V||-||-||Single Mode||Ø5.6 mm, SM Pigtail|
|LP1550-SAD2||1550 nm||2.0 mW||40 mA||1.0 V||-||-||Single Frequency||Ø5.6 mm, SM Pigtail|
|L1550P5DFB||1550 nm||5 mW||20 mA||1.1 V||8°||10°||Single Frequency||Ø5.6 mm|
|ML925B45F||1550 nm||5 mW||30 mA||1.1 V||25°||30°||Single Mode||Ø5.6 mm|
|SFL1550S||1550 nm||40 mW||300 mA||1.5 V||-||-||Single Frequency||SM Butterfly|
|SFL1550P||1550 nm||40 mW||300 mA||1.5 V||-||-||Single Frequency||PM Butterfly|
|LPSC-1550-FC||1550 nm||50 mW||250 mA||2 V||-||-||Single Mode||Ø5.6 mm, SM Pigtail|
|FPL1009S||1550 nm||100 mW||400 mA||1.4 V||-||-||Single Mode||SM Butterfly|
|FPL1009P||1550 nm||100 mW||400 mA||1.4 V||-||-||Single Mode||PM Butterfly|
|FPL1001C||1550 nm||150 mW||400 mA||1.4 V||18°||31°||Single Mode||Chip on Submount|
|FPL1055T||1550 nm||300 mW (Pulsed)||750 mA||2 V||15°||28°||Single Mode||Ø5.6 mm|
|FPL1055C||1550 nm||300 mW (Pulsed)||750 mA||2 V||15°||28°||Single Mode||Chip on Submount|
|SFL1620S||1620 nm||40 mW||300 mA||1.5 V||-||-||Single Frequency||SM Butterfly|
|LPSC-1625-FC||1625 nm||50 mW||350 mA||1.5 V||-||-||Single Mode||Ø5.6 mm, SM Pigtail|
|FPL1054S||1625 nm||80 mW||400 mA||1.7 V||-||-||Single Mode||SM Butterfly|
|FPL1054P||1625 nm||80 mW||400 mA||1.7 V||-||-||Single Mode||PM Butterfly|
|FPL1054C||1625 nm||250 mW (Pulsed)||750 mA||2 V||15°||28°||Single Mode||Chip on Submount|
|FPL1054T||1625 nm||250 mW (Pulsed)||750 mA||2 V||15°||28°||Single Mode||Ø5.6 mm|
|FPL1059S||1650 nm||80 mW||400 mA||1.7 V||-||-||Single Mode||SM Butterfly|
|FPL1059P||1650 nm||80 mW||400 mA||1.7 V||-||-||Single Mode||PM Butterfly|
|FPL1059C||1650 nm||225 mW (Pulsed)||750 mA||2 V||15°||28°||Single Mode||Chip on Submount|
|FPL1059T||1650 nm||225 mW (Pulsed)||750 mA||2 V||15°||28°||Single Mode||Ø5.6 mm|
|FPL1940S||1940 nm||15 mW||400 mA||2 V||-||-||Single Mode||SM Butterfly|
|FPL2000S||2 µm||15 mW||400 mA||2 V||-||-||Single Mode||SM Butterfly|
|FPL2000C||2 µm||30 mW||400 mA||5.2 V||8°||19°||Single Mode||Chip on Submount|
|FPL2000CM||2 µm||30 mW||400 mA||2 V||-||-||Single Mode||C-Mount|
|IF3300CM2||3.30 µm (FP)||20 mW||480 mA||2.5 V||35°||65°||Single Mode||Two-Tab C-Mount|
|IF3420CM2||3.42 µm (FP)||30 mW||510 mA||2.6 V||30°||45°||Single Mode||Two-Tab C-Mount|
|IF3800CM2||3.80 µm (FP)||30 mW||550 mA||2.5 V||40°||60°||Single Mode||Two-Tab C-Mount|
|QD4500CM1||4.00 - 5.00 µm (DFB)||40 mW||<500 mA||10.5 V||30°||40°||Single Frequency||Two-Tab C-Mount|
|QF4050CM1||4.05 µm (FP)||150 mW||1030 mA||12.5 V||28°||47°||Single Mode||Two-Tab C-Mount|
|QF4400CM1||4.40 µm (FP)||500 mW||1020 mA||10.7 V||26°||53°||Single Mode||Two-Tab C-Mount|
|QD4580CM1||4.54 - 4.62 µm (DFB)||40 mW||<600 mA||10.5 V||50°||30°||Single Frequency||Two-Tab C-Mount|
|QF4550CM1||4.55 µm (FP)||450 mW||900 mA||10.5 V||30°||55°||Single Mode||Two-Tab C-Mount|
|QF4800CM1||4.80 µm (FP)||500 mW||850 mA||15.5 V||33°||53°||Single Mode||Two-Tab C-Mount|
|QD5500CM1||5.00 - 8.00 µm (DFB)||40 mW||<700 mA||9.5 V||30 °||45 °||Single Frequency||Two-Tab C-Mount|
|QD5250CM1||5.20 - 5.30 µm (DFB)||120 mW||<660 mA||10.2 V||41°||52°||Single Frequency||Two-Tab C-Mount|
|QF5300CM1||5.30 µm (FP)||150 mW||1200 mA||9.0 V||30°||55°||Single Mode||Two-Tab C-Mount|
|QD6500CM1||6.00 - 7.00 µm (DFB)||40 mW||<650 mA||10 V||35 °||50 °||Single Frequency||Two-Tab C-Mount|
|QF7200CM1||7.20 µm (FP)||250 mW||1300 mA||8.5 V||35°||65°||Single Mode||Two-Tab C-Mount|
|QD7500CM1||7.00 - 8.00 µm (DFB)||40 mW||<600 mA||10 V||40°||50°||Single Frequency||Two-Tab C-Mount|
|QD7500DM1||7.00 - 8.00 µm (DFB)||100 mW||<600 mA||11.5 V||40°||55°||Single Frequency||D-Mount|
|QF7700CM1||7.70 µm (FP)||250 mW||1100 mA||7.8 V||37°||65°||Single Mode||Two-Tab C-Mount|
|QD7950CM1||7.90 - 8.00 µm (DFB)||100 mW||<1000 mA||9.5 V||55°||70°||Single Frequency||Two-Tab C-Mount|
|QD8050CM1||8.00 - 8.10 µm (DFB)||100 mW||<1000 mA||9.5 V||55°||70°||Single Frequency||Two-Tab C-Mount|
|QD8500CM1||8.00 - 9.00 µm (DFB)||100 mW||<900 mA||9.5 V||40 °||55 °||Single Frequency||Two-Tab C-Mount|
|QD8500HHLH||8.00 - 9.00 µm (DFB)||100 mW||<600 mA||10.2 V||-||-||Single Frequency||Horizontal HHL|
|QF8350CM1||8.55 µm (FP)||300 mW||1750 mA||8.5 V||55°||70°||Single Mode||Two-Tab C-Mount|
|QD8650CM1||8.60 - 8.70 µm (DFB)||50 mW||<900 mA||9.5 V||55°||70°||Single Frequency||Two-Tab C-Mount|
|QD9500CM1||9.00 - 10.00 µm (DFB)||60 mW||<800 mA||9.5 V||40°||55°||Single Frequency||Two-Tab C-Mount|
|QD9500HHLH||9.00 - 10.00 µm (DFB)||100 mW||<600 mA||10.2 V||-||-||Single Frequency||Horizontal HHL|
|QF9150CM1||9.15 µm (FP)||180 mW||1500 mA||8.4 V||40°||65°||Single Mode||Two-Tab C-Mount|
|QF9550CM1||9.55 µm (FP)||80 mW||1500 mA||7.8 V||35°||60°||Single Mode||Two-Tab C-Mount|
|QD9550CM1||9.50 - 9.60 µm (DFB)||40 mW||<1350 mA||9.5 V||28°||60°||Single Frequency||Two-Tab C-Mount|
|QD10500CM1||10.00 - 11.00 µm (DFB)||40 mW||<600 mA||10 V||40°||55°||Single Frequency||Two-Tab C-Mount|
|The rows shaded green above denote single-frequency lasers.|